Milli-Q® IQ 7000 specifications
|Ultrapure, Type 1 water specifications1 (from Q-POD® dispenser)|
|Resistivity at 25 °C2||18. 2 MΩ·cm|
|TOC3||≤ 2 ppb (µg/L)|
|Particles4||No particles with size > 0.22 µm|
|Bacteria||< 0.01 CFU/mL4,5
< 0.005 CFU/mL6
|Pyrogens (endotoxins)5||< 0.001 EU/mL|
|RNases5||< 1 pg/mL|
|DNases5||< 5 pg/mL|
|Proteases5||< 0.15 µg/mL|
|Flow rate||0.05 – 2 L/min|
|1 These values are typical and may vary depending on the nature and concentration of contaminants in the feed water.
2 Resistivity can also be displayed non-temperature-compensated as required by USP.
3 In the appropriate operating conditions.
4 With Millipak® and Millipak® Gold
5 With Biopak®
6 With Millipak® Gold installed and used in a laminar flow hood
International regulation requirements
Declaration of CE and cUL Conformity:
The Milli-Q® IQ 7000 system has been tested by an independent and accredited company for compliance with CE directives related to safety and electromagnetic compatibility. The report can be consulted on demand
. In addition, the Milli-Q® IQ 7000 system is built using components and practices recommended by UL and has been cUL marked
Ex-vivo studies with human platelets indicated that sildenafil potentiates the antiaggregatory effects of sodium nitroprusside; these effects did not result in a significant effect on bleeding time in healthy volunteers.include its less invasive nature. The disadvantages include cialis online.
. The registration can be verified on the UL web site:http://www.ul.com.